[Prevention of tracheal lesions by rhythmic insufflation of the intubation catheter ballone].
The prevention of tracheal lesions due to the ballons of the intubation catheters depends on the correction of several factors concerning: -the material used (polyvinyl is better tolerated than rubber); -the shape (the sleeve-shape balloons are less traumatic); -the pressure of use, which should remain moderate; -the permanence of the compression (one should use intermittent filling of the ballon). A device designed by the authors and which is described here, permits intermittent insufflation of the ballons of the endotracheal catheters, by adaptation of certain apparatus of artificial ventilation, type LOGIC and SF4, has given excellent results.
[Prevention of tracheal lesions by rhythmic insufflation of the intubation catheter ballone]. The prevention of tracheal lesions due to the ballons of the intubation catheters depends on the correction of several factors concerning: -the material used (polyvinyl is better tolerated than rubber); -the shape (the sleeve-shape balloons are less traumatic); -the pressure of use, which should remain moderate; -the permanence of the compression (one should use intermittent filling of the ballon). A device designed by the authors and which is described here, permits intermittent insufflation of the ballons of the endotracheal catheters, by adaptation of certain apparatus of artificial ventilation, type LOGIC and SF4, has given excellent results.